### Days Postings

December 2020 M T W T F S S 1 2 3 4 5 6 7 8 9 10 11 12 13 14 15 16 17 18 19 20 21 22 23 24 25 26 27 28 29 30 31 ### Blog Series

### Subscribe to Blog via Email

### Copyright Notice

© Mark Biegert and Math Encounters, 2020. Publication of this material without express and written permission from this blog’s author and/or owner is strictly prohibited. Excerpts and links may be used, provided that full and clear credit is given to Mark Biegert and Math Encounters with appropriate and specific direction to the original content.

### Disclaimer

All content provided on the mathscinotes.com blog is for informational purposes only. The owner of this blog makes no representations as to the accuracy or completeness of any information on this site or found by following any link on this site. The owner of mathscinotes.com will not be liable for any errors or omissions in this information nor for the availability of this information. The owner will not be liable for any losses, injuries, or damages from the display or use of this information.

# Category Archives: Metrology

## Measuring Countersink Angle Using Gage Balls

This post will cover how to measure a countersink angle using gage balls. Figure 1 shows how a countersink is normally specified on an engineering drawing. I frequently use countersinks in my wood and metal working hobbies. In addition, using gage balls to measure the countersink angle provides a good example of how to apply basic geometry concepts to a practical problem. I use this example in my role as a volunteer adult math tutor at our local library. Continue reading

Posted in General Mathematics, Metrology
Leave a comment

## Shotgun Bore Diameter Math

I have been doing some metalwork lately that involves using units of "gauge". You will find the term gauge used in the measurement of wire, metal thickness, and pipe bore diameter. This quaint, but confusing, measurement system is slowly falling out of favor (example, sheet metal thickness gauge). Continue reading

Posted in Metrology
3 Comments

## Measuring a Chamfer Angle Using Gage Balls

One metrology operation I have had to perform a number of times is measuring a chamfer angle precisely – Figure 1 shows today's example. Many items are chamfered – even in electronics. For example, edge connectors on printed circuit boards often need to be chamfered to ensure that they do not damage the connectors they are being inserted into. Continue reading

## Tapered Side Angle Measurement

I thought I was done with my metrology review when I encountered an excellent set of discussions at the Hobby-Machinist web site. They advertise themselves as "The Friendly Machinist Forum" and all signs indicate that is true. In addition to excellent tutorial discussion, there are some excellent metrology discussions on that site, and I want to document a few of the examples that are shown there. Continue reading

Posted in Metrology
2 Comments

## Another Angle Measurement Using Roller Gages Plus Error Analysis Example

While I covered angle measurement in a previous post, that approach can be difficult to apply for acute angles. The approach presented in this post works well for acute angles, but will not work for obtuse angles.

As part of this post, I will also demonstrate how to perform a tolerance analysis on this approach. The tolerance analysis is important in understanding the level of accuracy required in your linear measurements to achieve the desired angle accuracy. Continue reading

## Measuring Round-Over Radius Using a Roller Gage

This blog post shows how to measure the radius on a rounded corner (a.k.a. bullnose) using a roller gage. I saw this method being used on this web page, and I wanted to document it here for future reference. Continue reading

## Bore Diameter Measurement Using Gage Balls

I am continuing to work through the metrology examples on this web page as part of junior machinist self-training. Today's technique shows how to use gage balls to measure the bore diameter of a cylinder (Figure 1). You can measure a bore diameter using a micrometer, but I have concerns that I might be measuring along a chord instead of a diameter – this error would result in too small of a result. The gage ball approach should eliminate that type of error. Continue reading

Posted in Metrology
4 Comments

## Angle Measurement Using Roller Gages

I am continuing to work through some basic metrology examples – today's example uses roller gages to measure the angle of a drilled hole (Figure 1). The technique discussed here uses two roller gages and a plug. The plug must fit the hole snugly (i.e. no backlash) as it will provide the surface that we will be measuring. Using this approach assumes that you need a very accurate measurement of a hole's angle as rough measurements can be made using a protractor. Continue reading

Posted in Metrology
2 Comments

## Radius Measurement Using Roller Gages

This post will demonstrate how to measure the radius of an arc using two roller gages. While I am a very amateur machinist, I have on occasion needed to measure the radius of an arc (i.e. partial circle) and have not been sure how to approach that measurement. It turns out to be simple given two equal diameter roller gages and a surface plate. You can determine by taking one measurement and knowing the roller gage diameter. Continue reading

Posted in Construction, Geometry, Metrology
3 Comments

## Measuring Countersink Diameter Using Gage Balls

I am still working through some examples of using gage balls for machine shop work. The following reference on Google Books has great information on using gage balls (Figure 1) in measuring the characteristics of a countersink and I will be working through the presentations there. These are good, practical applications of high-school geometry. Continue reading

Posted in Construction, Geometry, Metrology
15 Comments